Medical Diagnostics 200mA X-Ray Machine.
Single-table / Single-tube
Rotary anode X-ray tube unti and tangential annular tube head.
Single-phase full wave rectification high-voltage generator.
Power voltage (V) radiograph kilovolt (KV) .Step-less control electric mechanism.
Filament stabilizer for X-ray tube and space charge complementor
Radiograph volume KV mA and S. grading and interlock protection (for more accurately controlling time )
HT primary, the zero controlled circuit of silicon-controlled rectifier of heavy power
Radiographic table can move in length and breath.
The radiographic table,column and vibrating grid are in a whole without ceiling and floor rails bottom.
|Demand of power supply
|380V/220V ±10% 50HZ ±0.5HZ
|0.75 Ω(380V) 0.25Ω(220V)
|±10% at 380V,220±10% AT 220V being adjustable continuously
|50-100KV being adjustable continuously
|Small focus:50mA 100mA,Big focus:50mA 100mA 150mA 200mA
|0.04-6.35s 23grades together grading according to R10 coefficient
|Max DC output voltage
|Max DC output currency
|Small focus:1x1mm,Big focus:2x2mm
|width xheight of bed surface
|2000mm x 665mm x700mm
|Moving range of bed surface
|640mm in lengh,200mmin breadth
|Ray-fillter of bed
|Traval in length 500mm
|Grid density N=28
|Convergence distance f.=100cm
|Grid ratio r=8