Medical Diagnostics 200mA X-Ray Machine.
Single-table / Single-tube
Rotary anode X-ray tube unti and tangential annular tube head.
Single-phase full wave rectification high-voltage generator.
Power voltage (V) radiograph kilovolt (KV) .Step-less control electric mechanism.
Filament stabilizer for X-ray tube and space charge complementor
Radiograph volume KV mA and S. grading and interlock protection (for more accurately controlling time )
HT primary, the zero controlled circuit of silicon-controlled rectifier of heavy power
Radiographic table can move in length and breath.
The radiographic table,column and vibrating grid are in a whole without ceiling and floor rails bottom.
|Demand of power supply||Capacity||30KVA(Two-phase 30KVA)|
|Voltage frequency||380V/220V ±10% 50HZ ±0.5HZ|
|Internal resistance||0.75 Ω(380V) 0.25Ω(220V)|
|Power suppy||Adjustable range||±10% at 380V,220±10% AT 220V being adjustable continuously|
|Radiograph||Voltage||50-100KV being adjustable continuously|
|Current||Small focus:50mA 100mA,Big focus:50mA 100mA 150mA 200mA|
|Time||0.04-6.35s 23grades together grading according to R10 coefficient|
|Max DC output voltage||100KV|
|Max DC output currency||200mA|
|Focus||Small focus:1x1mm,Big focus:2x2mm|
|Radiographic table||width xheight of bed surface||2000mm x 665mm x700mm|
|Moving range of bed surface||640mm in lengh,200mmin breadth|
|Ray-fillter of bed||Traval in length 500mm|
|Grid density N=28|
|Convergence distance f.=100cm|
|Grid ratio r=8|